Immunoglobulins or antibodies are proteins that are involved in the immune response. That makes them interesting drug targets. This application note shows the detection of the immunoglobulin IgG using an AlphaLISA® assay performed on BMG LABTECH's PH more
Molecular Probes® NanoOrange® Assay Performed on BMG LABTECH FLUOstar OPTIMA Microplate ReaderTracey Madgett, University of the West of England, Bristol
Faculty of Health and Life Sciences, Frenchay Campus, Coldharbour Lane, Bristol, BS16 1QY, United Kingdom
- NanoOrange® Assay for protein quantitation performed on the FLUOstar OPTIMA
- High and small concentration range evaluated
- Kit can easily be adapted for high throughput measurements
Fig. 1: Excitation and emission spectra of the NanoOrange® reagent. Copyright Invitrogen Corporation. Used with permission.
The field of proteomics has expanded dramatically in recent years with research on a whole host of organisms. Techniques such as two dimensional difference gel electrophoresis (2D DIGE) require the accurate quantitation of protein, as up to three labelled samples can be loaded on a single gel for comparison. Assay methods for determining protein quantitation include absorbance at 280 nm(1), the Bradford assay(2), Lowry assay(3), BCA method(4) and more sensitive assays such as Fluoroprofile® (Sigma-Aldrich) and NanoOrange® (Molecular Probes®)5(). The measurement of solution absorbance at 280 nm (A280) has problems with variability between samples and interference from nucleic acids and other contaminants. Detergents and reducing agents can cause problems with the other assays mentioned above – these agents are present in samples for 2D gel electrophoresis. Fluorescent methods are more sensitive for quantitating proteins than absorbance methods. The NanoOrange® protein quantitation assay from Molecular Probes® is a highly sensitive assay with the useful range being between 100 ng/mL and 10 µg/mL for fluorescence-based microplate readers. The spectra of the NanoOrange® reagent is given in Figure 1.
The sensitivity persists over 6 hours and there is low protein-to-protein variability. Tolerance levels are as high as 100 mM for reducing agents such as DTT and β-mercaptoethanol and 0.01% for detergents such as SDS. This application note investigates a method for using the NanoOrange® assay on the BMG LABTECH FLUOstar OPTIMA microplate reader.
Materials and Methods
The following materials were supplied by the manufacturers as detailed:
- NanoOrange® Protein Quantitation Kit (Catalogue number N6666) (Molecular Probes®, Invitrogen, Paisley, United Kingdom)
- MJ Research PTC200 Peltier Thermal Cycler (MJ Research, now distributed by Bio-Rad, Hemel Hempsted, United Kingdom)
- FLUOstar OPTIMA Microplate Reader (Part number 413-101) (BMG LABTECH, Aylesbury, United Kingdom, Figure 2)
- Microplates, black 96 well (Catalogue number 655076) (Greiner Bio-One, Stonehouse, United Kingdom)
- General laboratory consumables included pipette tips and microcentrifuge tubes
A description of the preparation for microplate assays has been noted earlier (http://probes.invitrogen.com/media/pis/mp06666.pdf).
- NanoOrange® Protein Quantitation reagent A
- NanoOrange® Protein Quantitation diluent B
- Bovine Serum Albumin (BSA) Standard (2 mg/mL)
The working NanoOrange® reagent was made by diluting reagent A 500-fold in a 1:10 dilution of diluent B. A stock BSA solution at 10 µg/mL was made by diluting 1:200 in working NanoOrange® reagent. A serial dilution of the stock BSA solution was performed, yielding further concentrations ranging from 0.2 µg/mL to 10 µg/mL. The blank solution was working NanoOrange® reagent alone. Dilutions of samples were made in working NanoOrange® reagent (for example, 1:100, 1:500, 1:1000). Following denaturation of the standards and samples, 100 µL of each was placed into a microplate for measurement. The prepared 96-well plate was inserted into the instrument and read in fluorescence mode with the following parameters:
|Excitation filter:||485 BP12|
|Gain:||plate assessed, well with the highest
|Optics:||2mm combination optic|
|Number of cycles:||1|
|Number of flashes per well:||10|
The data was evaluated using the BMG LABTECH FLUOstar evaluation package. The average value of the blank measurement was subtracted from the measurements and the standard curve was plotted.
Fig. 3: BSA standard curve obtained with the NanoOrange® Protein Quantitation Kit. A BSA concentration range of 0.2-10 µg/mL is used. A gain of 1967 was optimized on the highest protein concentration.
Results and Discussion
A four parameter fit was performed on the data which yielded a very high R-value of 0.9998. (Figure 3).
The small concentration range was also measured separately with optimized gain on the highest protein concentration (Figure 4).
Fig. 4: Small concentration range BSA standard curve obtained with the NanoOrange® Protein Quantitation Kit. A gain of 2721 was optimized on the highest protein concentration (12 replicates).
A linear relationship between protein concentration and fluorescence units is obtained indicating that small concentrations of protein in samples can be determined using the FLUOstar OPTIMA. Protein concentrations below 200 ng/mL can be determined with the help of the 3 mm fluorescence optic.
The NanoOrange® Protein Quantitation Kit can be easily adapted for use with a fluorescent microplate reader such as the FLUOstar OPTIMA and used in a high throughput manner. NanoOrange® can also be performed on other BMG LABTECH microplate readers, including the FLUOstar and POLARstar Omegas, as well as the NOVOstar and PHERAstar.
The FLUOstar OPTIMA is a flexible multidetection microplate reader that has four different measurement modes: fluorescence intensity, time-resolved fluorescence, luminescence and absorbance. The instrument allows measurement of endpoint or slow and high kinetics at a user-defined temperature and can be easily equipped with injectors.
 Scopes, R.K., (1987), Protein Purification, Principles and Practice, 2nd Edition, Springer-Verlag
 Bradford, M.M., (1976), A rapid and sensitive method for the quantitation of microgram quantities of protein utilizing the principle of protein-dye binding. Anal. Biochem. 72; 248-254.
 Lowry, O.H., Rosebrough, N.J., Farr, A.L. & Randall, R.J., (1951), Protein measurement with the Folin phenol reagent. J. Biol. Chem. 193; 265-275.
 Smith, P.K., Krohn, R.I., Hermanson, G.T., Mallia, A.K., Gartner, F.H., Provenzano, M.D., Fujimoto, E.K., Goeke, N.M., Olson, B.J. & Klenk, D.C., (1985), Measurement of protein using bicinchoninic acid. Anal. Biochem. 150; 76-85. Erratum in: Anal. Biochem. (1987) 163; 279.
 “NanoOrange® Protein Quantitation Kit, Product Information, Molecular Probes®, http://probes.invitrogen.com/media/pis/mp06666.pdf
- microplate readers
- Bmg Labtech
- gel electrophoresis
- United Kingdom
- serum albumin
- protein purification
- pipette tips
- Life Sciences
- Greiner Bio-One
The potential for cardiotoxic side-effects continues to challenge the development of small molecule based therapies. These intolerable side-effects are often precipitated by drug-induced long QT syndrome (LQT), which is often linked to blocking the h more
Nicotinamide Adenine Dinucleotide (NAD+), reduced nicotinamide adenine dinucleotide (NADH), nicotinamide adenine dinucleotide phosphate (NADP+) and reduced nicotinamide adenine dinucleotide phosphate (NADPH) have been known to play vital roles in ene more
The CLARIOstar® from BMG LABTECH is a powerful multimode microplate reader with Advanced LVF Monochromators™, highly sensitive filters, and an ultra-fast spectrometer. It is a modular more
Robert J. Lefkowitz from Duke University, along with Brian Kobilka from Stanford University, were just awarded the 2012 Nobel Prize in Chemistry for their research and discovery of the receptor protein that binds to adrenaline, that is the beta-adrenergic receptor. A receptor protein crosse ... more
One of the first uses of the microscope by Louis Pasteur was to determine why beer and wine soured; he later hypothesized the Germ Theory of Disease. It’s in that same spirit that BMG LABTECH congratulates Dr. Gary Spedding for his proposal to use the new SPECTROstar Nano to modernize beer ... more
The ALSSA - Analytical & Life Science Systems Association - Board of Directors recently approved applications for membership submitted by BMG LABTECH, Inc. Ron Earp PhD, President, will serve as the official representative to ALSSA and E.J. Dell PhD, Business Development and Applications Sc ... more
- 1Microwave freeze drying of fruits & vegetables
- 2UV/Vis/NIR Spectrometry for the Film Thickness Determination of Transparent Films and Layers
- 3Optimization of Transfection Conditions for the Human HL-60 Promyelocytic Leukemic T-Cell Line Using FuGENE® HD Transfection Reagent
- 4Do you see the complete picture?
- 5Dynamic Light Scattering - an Important Tool for Protein Crystallography and Nanotechnology
- 6Fugene® HD Transfection Reagent: Superior Performance for Challenging Expression Studies
- 7Real-Time PCR Quality Control for Gene Expression
- 8Looxster® - Enrichment and isolation of bacterial DNA using Pureprove® Technology
- 9Field-Flow Fractionation - The Universal Separation Principle for Particle and Macromolecule Characterization
- 10Automated Dynamic Light Scattering