Polyethylene and Polypropylene are so-called Polyolefins. They are produced in large amounts every year. Depending on the process and catalyst used linear or branched polymers can be produced. In case of e. g. Polyethylene there are at least three polymer types: HDPE (High Density Polytheyl ... more
Measurement of Nano particles and ProteinsUlf Nobbmann, Biophysical Characterization, Malvern Instruments Ltd., UK
Renate Hessemann, Marketing Manager Europe, Malvern Instruments, Germany
Colloid emulsions in their native environment require particle size determination in high concentration since there are concerns about changes in sample morphology upon dilution eg break up of aggregates. For proteins and other samples with very low concentration and for weakly scattering samples high sensitivity is required.
New technologies offer both.
Dynamic Light Scattering (DLS) is a powerful technique for determining the size of sub-micron particles. Conventional instrumentation is limited in terms of the maximum concentration of samples that can be analysed because of multiple scattering effects. Non-invasive back scatter (NIBS) technology not only increases the concentration limits at which DLS can be successfully applied, but also increases the sensitivity of the technique.
Dynamic light scattering
Dynamic light scattering (DLS) is a non-invasive technique for measuring particle size, typically in the sub-micron size range. Particles in suspension undergo random Brownian motion. If these particles are illuminated with a laser beam, the laser light is scattered. The intensity of the scattered light detected at a particular angle fluctuates at a rate that is dependent upon the particle diffusion speed, which in turn is governed by particle size. Particle size data can therefore be generated from an analysis of the fluctuations in scattered light intensity
Concentration limits of DLS
In the past 20 years, DLS has become a routine tool for the measurement of particles less than a micron. However, conventional DLS has its limitations. The sample concentration must be high enough to ensure an adequate signal, but the risk of erroneous results due to multiple scattering (light scattered by one particle undergoing scattering by another) means there are restrictions on the concentration range for which measurements are valid.
NIBs extends the concentration limits
The use of patented noninvasive backscatter technology has overcome these limitations. NIBS is a dynamic light scattering technique incorporating an optical configuration that maximizes the detection of scattered light while maintaining signal quality. This provides the high sensitivity needed for measurement of the size and molecular weight of molecules smaller than 1000 D. It also enables measurement at extremely high concentrations. The use of backscattering, rather than more typically detecting scattered light at a 90o angle, improves the sensitivity and at the same time ensures the smallest possible interference from multiple scattering. Previous backscattering techniques have suffered from drawbacks that include the need for close contact between sample and detector optics, necessitating frequent cleaning of both the measurement cell and the detector. Because NIBS is a non contact technique, cleaning is not necessary.
The range of sample concentrations that can be analyzed successfully is extended by changing the measurement position within the cuvette. This is achieved by moving the focussing lens. For small particles, or samples at low concentrations, it is beneficial to maximize the amount of scattering from the sample and hence a measurement position towards the centre of the cuvette is most effective. Large particles, or samples at high concentrations, scatter much more light and therefore measuring closer to the cuvette wall is preferable as this reduces the chance of multiple scattering.
DLS for protein characterization
One of the most time consuming steps is still the actual crystallisation, the search for the conditions under which the protein under investigation will crystallize. As the screening involves a plethora of various buffer conditions and protein amounts are often limited it has become wide-spread practice to check the suitability of the starting material. A simple light scattering experiment will tell the size and the polydispersity, the non-homogeneity of the starting sample. The measurement is quick, and requires only small volumes, with the intact sample being available for further analysis.
What information does one get out of the technique?
The size itself can be linked to the molecular weight of the protein. While there may be unusual shape effects, many proteins tend to behave like relatively globular molecules. And these may then be expected to behave according to the Mark-Houwink relation: the measured particle size is related to the molecular weight through a power law. When encountering an ‘unknown’ protein, it is a simple matter of comparing its measured size to the expected estimated molecular weight. Thus, the size can predict the oligomeric state of the protein in solution. As the measured size gives an impression of the molecules as it is present in the sample under the current conditions, it provides insight into the actual oligomeric configuration. This, however, requires a reasonable data quality. Very polydisperse samples are not suitable for such advanced data interpretation.
The polydispersity is the width of the size distribution. When many different particle species are present in a measuring volume, the width of the size distribution will turn out to be larger than when all particles are of the same size.
In real life, there seems to be a “natural polydispersity” due to constant interchange with solvent layer molecules and some molecular flexibility. However, when different particle species, such as dimers, trimers, oligomers are present then the polydispersity is markedly higher than for monodisperse single-meric solutions.
The relative polydispersity expressed in percent of the half width of the peak divided by the peak mean in the particle size distribution can vary from a few to a hundred percent. Many proteins show polydispersities below 20% for single species, 20-30% for oligo-species (monomer-dimer, or monomer-tetramer), and above 30% when forming a wide range of different oligomeric states in the buffer under question.
If the question is solubility at different buffer compositions then light scattering is the fastest answer. It provides the particle size (which gives an estimate of the oligomeric state of the protein) and the polydispersity (which shows the homogeneity of the distribution in solution). The ease and speed of the technique has shown advantages in protein crystallization, stability analysis, thermal properties, degradation, self-assembly, in short characterization in solution conditions.
Zetasizer Nano - optimized for Protein characterization
Malvern Instruments’ Zetasizer Nano S combined static and dynamic light scattering instrument is optimized for the characterization of proteins in solution prior to crystallization.
This compact, easy to use system is designed for the rapid delivery of accurate and extensive information that can assist both in the screening of appropriate conditions for protein crystallization and in determining the likelihood of crystals being suitable for structure determination. Not only does the Zetasizer Nano automatically optimize all instrument settings for each sample but custom data reports and graphical data presentations make interpretation easier than ever before.
The Zetasizer Nano enables researchers to detect and quantify aggregation, determine the second virial coefficient to find the “crystallisation sweet spot”, and quantify sample polydispersity to increase the likelihood of successful crystallization.
In addition it enables users to study the effect of temperature on monodispersity, and offers the ability to automate temperature studies, including melting point and thermal denaturation determinations.
It also allows estimation of prolate and oblate axial ratios and Perrin factor, as well as measurement of hydrodynamic diameter and absolute molecular weight.
Size measurements of proteins as small as 0.6 nm and 400 Da can be made in their native environments. As little as 12 microliters of sample is required and the sample is recoverable.
- Sweet Spot
- stability analysis
Dynamic Light Scattering (DLS) is an established technique in the field of protein crystallography. DLS is used to measure hydrodynamic sizes, polydispersities and aggregation effects of protein samples. These are important parameters for the crystallisation of proteins. DLS is also used in ... more
Fully automated particle size measurement technology integrated with existing plant control platforms for ease of use has the potential to deliver numerous benefits. At plants where a switch to on-line analysis has been made it is frequently easier or possible to: more
Malvern Instruments Ltd has announced that it has completed the acquisition of MicroCal from GE Healthcare Life Sciences and has also purchased the Archimedes particle characterization system from Affinity Biosensors LLC (Santa Barbara, CA). Both of these developments extend Malvern’s portf ... more
Malvern Instruments Ltd. announced that it has entered into an agreement with GE Healthcare Life Sciences to acquire the company’s MicroCal™ microcalorimetry business. Financial terms were not disclosed. The acquisition, which is subject to regulatory approvals, is expected to complete i ... more
NanoSight Ltd have been acquired by Malvern Instruments Ltd (Malvern, UK). The acquisition was completed on 30th September 2013. NanoSight has developed and commercialized a unique and patented technology, Nanoparticle Tracking Analysis (NTA). The company’s instrumentation and software enab ... more
This seminar is aimed at chemists, biologists and engineers who are less experienced in rheology or who would like to like to expand their rheology knowledge. It will provide an introduction to rheology, detail how rheology is relevant in various applications and other techniques, and pre ... more
Learn More – Review theory behind laser diffraction particle sizing and how to apply it successfully to your work. Explore alternative methods like particle image analysis as well. See More – Meet technology experts and see our latest software and instrument developments. more
- 1Microwave freeze drying of fruits & vegetables
- 2UV/Vis/NIR Spectrometry for the Film Thickness Determination of Transparent Films and Layers
- 3Fugene® HD Transfection Reagent: Superior Performance for Challenging Expression Studies
- 4Optimization of Transfection Conditions for the Human HL-60 Promyelocytic Leukemic T-Cell Line Using FuGENE® HD Transfection Reagent
- 5Real-Time PCR Quality Control for Gene Expression
- 6Looxster® - Enrichment and isolation of bacterial DNA using Pureprove® Technology
- 7Good Reasons to Use ATR Libraries
- 8Automated Dynamic Light Scattering
- 9LOOXSTER® concentrates bacterial DNA
- 10Superior Transfection of Human Embryonic Stem Cells with FuGENE® HD Transfection Reagent